How to linearize arrayed data
It is possible to open an arrayed data in Data Slate and to use the "Linearize" function in order to arrange all the 1D spectra side by side. This is useful if you measure pulse width or diffusion coefficient, for instance. In this example, we will show how to measure pulse width.
Select Expansion – Linearize.
∗ Refer to User’s Manual on how to collect arrayed data (i.e. pseudo 2D data) on JEOL NMR spectrometers. In this example, obs_width_90 was varied from 1 μs to 19 μs (step 0.5 μs). All the 1D spectra were stored as a single pseudo-2D data file.
⇑ The plot of signal intensity vs. pulse width consists of 37 one-dimensional 13C spectra arranged side by side.
⇑ To simplify the plot, it might be convenient to zoom in on a specific peak (e.g. 38.6 ppm) rather than a group of peaks.
- If you want to see a printed version,
please click this PDF file.